C-reactive protein is considered one of the best measures of the acute-phase response to an infectious disease or other cause of tissue damage and inflammation. It is used to correct the iron status measures, which are affected by inflammation. It can also be used to measure the body’s response to inflammation from chronic conditions, such as arthritis, and environmental exposures to agents such as tobacco smoke.
Participants aged 3 to 5 year and 12 and older.
Blood specimens are processed, stored and shipped to University of Washington, Seattle, WA. Detailed specimen collection and processing instructions are discussed in the NHANES Laboratory/Medical Technologists Procedures Manual (LPM).
This method quantified CRP by latex-enhanced nephelometry. Particle-enhanced assays were based on the reaction between a soluble analyte and the corresponding antigen or antibody bound to polystyrene particles. For the quantification of CRP, particles consisting of a polystyrene core and a hydrophilic shell were used to link anti-CRP antibodies covalently. A dilute solution of test sample was mixed with latex particles coated with mouse monoclonal anti-CRP antibodies. CRP present in the test sample forms an antigen antibody complex with the latex particles.
An automatic blank subtraction was performed. CRP concentrations were calculated by using a calibration curve. Data reduction of the signals was performed by using a storable logit-log function for the calibration curve performed data reduction of the signals. These assays were performed on a Behring Nephelometer for quantitative CRP determination.
There were no changes (from the previous 2 years of NHANES) to equipment, lab methods or lab site.
A detailed description of the laboratory method used can be found on the NHANES website.
See the NHANES web page link to determine “below detectable limit fill values” for this data. The lower detectable limit is .02.
Read the General Documentation of Laboratory Data file for detailed data processing and editing protocols. The analytical methods are described in the Analytic methodology section.
The NHANES quality control and quality assurance protocols (QA/QC) meet the 1988 Clinical Laboratory Improvement Act mandates. Detailed quality control and quality assurance instructions are discussed in the NHANES Laboratory/Medical Technologists Procedures Manual (LPM). Read the General Documentation of Laboratory Data file for detailed QA/QC protocols.
A detailed description of the quality assurance and quality control procedures can be found on the NHANES website.
The analysis of NHANES laboratory data must be conducted with the key survey design and basic demographic variables. The NHANES Household Questionnaire Data Files contain demographic data, health indicators, and other related information collected during household interviews. They also contain sample weights for these age groups. The phlebotomy file includes auxiliary information such as the conditions precluding venipuncture. The household questionnaire and phlebotomy files may be linked to the laboratory data file using the unique survey participant identifier SEQN.
The detection limit was constant in the data set. In cases, where the result was below the limit of detection, the value for that variable is the detection limit divided by the square root of two.
Exam sample weights should be used for analyses. Please refer to the Analytic Guidelines for further details on the use of sample weights and other analytic issues. The Analytic Guidelines are available on the NHANES website.
|Code or Value||Value Description||Count||Cumulative||Skip to Item|
|0.01 to 20||Range of Values||6809||6809|